Only the Endurant™ II stent graft system has the deep clinical experience and favorable clinical outcomes designed to treat both challenging and straightforward. The Endurant® II AAA Stent Graft System is designed to treat abdominal aortic aneurysms using a minimally invasive procedure called endovascular stent. The Endurant™ II/IIs AAA Stent Graft System used in combination with the Heli- FX™ EndoAnchor™ System is the first AAA short neck solution independent of.
|Published (Last):||1 September 2007|
|PDF File Size:||5.52 Mb|
|ePub File Size:||20.12 Mb|
|Price:||Free* [*Free Regsitration Required]|
Leverages the proven design of the leading Endurant II abdominal stent graft Expands anatomical customization options Offers a shorter ipsilateral leg for more flexible, targeted limb placement Features equal leg diameters to allow limbs to be used on either side Utilizes the existing delivery system allowing accurate placement and controlled deployment The Endurant IIs stent graft: Close Informativa per l’uso dei cookie. Si, sono un operatore sanitario Note: With an updated browser, you medtroniic have a better Medtronic website experience.
Your browser is out of date With an updated browser, you will have a better Medtronic website experience. Sei un operatore sanitario? Control the release of the suprarenal stent and anchor enduraht with the back-end thumb wheel. Federal USA law restricts this device to sale by mmedtronic on the order of a physician.
Related Searches Blood Glucometer Drug eluting Scaffold Irrigation pump Coronary stent Valve prosthesis Metdronic surgical power tool Medtronic patient monitor Electronic control unit Ball tube electrode Electrosurgical device Metal stent Pulse oximeter with capnograph Transportable ventilator Coagulation analyser Continuous positive airway pressure respirator Bone support Mdetronic stent Electrophysiologic monitoring catheter Interbody fusion cage Endutant patient monitor.
Suprarenal stent anchor pins provide secure fixation. Low-profile, hydrophilic delivery coating enhances access and trackability. Flexibility and Conformability — The Endurant II system conforms to a variety of anatomies including tortuous iliac arteries and highly angulated necks, and treats necks as short as 10mm with angulation up to and eneurant 60 degrees.
Open the catalog to page 2. Manuals and Technical Guides Instructions for Use Find this technical manual in the Medtronic Manual Libraryin the product labeling supplied with each device, or by calling Renal complications may occur: When placed within the aneurysm, the Endurant II Stent Graft provides a permanent, alternative conduit for blood flow within the patient’s vasculature.
There are two types of Endurant II delivery systems: Update my browser now.
Update my browser now. Optimal seal and fixation. The Endurant II Stent Graft System aorto-uni-iliac AUI stent graft is indicated for the endovascular treatment of infrarenal abdominal aortic or aortoiliac aneurysms in patients whose anatomy does not allow the use of a bifurcated stent graft. Please refer to the product Instructions for Use for details.
With an updated browser, you will have a better Medtronic website experience. Le informazioni che raccogliamo con i cookies sono informazioni non personali. medtronix
Endurant AAA Stent Graft System
It can be scanned safely in both 1. Medtronic utilizza dei cookies sui suoi siti web. Bench test data may not be indicative of clinical performance.
M-shaped proximal stents provide wall apposition and minimize in-folding. All patients should be advised that endovascular treatment requires medtrronic, regular follow-up to assess the health and the performance of the implanted endovascular stent graft.
Endurant II Stent Graft Systems
Aggiorna il mio browser adesso. When using the mm length bifurcated medtrknic graft, subtract 10 mm from Total Contralateral Covered Length with Bifurcated. Adverse Events Potential adverse events include arranged in alphabetical order: For more information, download the Endurant II product brochure and visit www.
Non Accetto termini e condizioni dei Cookies Accetto termini e condizioni dei Cookies.
R 2 As a result of emboli or a misplaced stent graft. Instructions for Use Find this technical manual in the Medtronic Manual Libraryin the product labeling supplied mrdtronic each device, or by calling